The Filta-Max module comprises multiple layers of open-cell foam discs that are compressed to create a filter with a nominal pore size to effectively trap Cryptosporidium and Giardia. After sampling, the filter is processed by releasing the compression of the foam discs and washing, enabling all captured material to be recovered. It is a simple yet extremely efficient system.
To sample, the filter module is placed in a reusable housing, which is attached to the required sampling source. A head pressure of 0.5 Bar (7.5 psi) is required to create a flow, and the recommended operating pressure is 5 Bar (75 psi). This pressure will create a flow rate of 4 liters per minute, which is particularly beneficial when sampling large volumes of water.
After filtration, the module is removed from the housing and placed in the IDEXX wash station. The foam discs are decompressed by removing a bolt in the module and then washed with an elution buffer using expansion/compression cycles.
The volume of buffer is then reduced to between 25 and 50 mL by concentrating through a membrane, negating the need for high-volume centrifugation.
Connect the housing to the required water source for sampling.
Remove the filter module from the housing.
Place the module in a wash station and wash with elution buffer.
Concentrate the wash buffer using a 3-µm membrane.
Repeat steps 3 and 4 to give a final volume of 25 mL.
Is Filta-Max listed in the Standing Committee of Analysts Blue Book?
Yes, Filta-Max is specified in the Microbiology of Drinking Water (MoDW) Part 14, 2010.
What volumes of water can be filtered using the Filta-Max system?
The system is efficient for both raw and finished water. The filter has been approved for sampling 50 liters of raw water. The Filta-Max system is also approved for sampling 1,000 liters of finished water.
What flow rates can be achieved with the Filta-Max system?
The Filta-Max system can operate at a flow rate of up to 4 liters per minute.
What are the recommended pressures for sampling?
A minimum pressure of 7.5 psi is required. 50 psi–75 psi is the recommended sampling pressure.
What are the shelf life and storage conditions for the Filta-Max filter module?
The shelf life is 18 months from the date of manufacture. Store module in a dry place (16°C–24°C), away from direct sunlight. Do not freeze. Do not refrigerate prior to use.
How should the filter be transported?
The filter should be transported either in the reusable housing or in a watertight resealable bag along with several milliliters of reagent-grade water.
What recoveries can be achieved with the Filta-Max system?
Recoveries are consistently in excess of the minimum required by the regulations.
How does the cost of the Filta-Max system compare with other filters on the market?
The Filta-Max system is approximately half the cost of other approved filters.
What is the variability of the method?
Variability is low. Relative standard deviations are up to 50% lower than other approved filters.
IDEXX Water has reference materials and approval documents to support the many products in our water portfolio.
IDEXX Technologies Ltd Units 1B & 1C, Newmarket Business Park, Studlands Park Avenue, Newmarket, Suffolk, CB87ER
Tel: 01638 676800